1. Industrial endoscope is a kind of precision instrument. If it is impacted by strong external force, the equipment may be damaged. Therefore, when taking out the instrument from the instrument box and carrying out the operation, we must pay attention to handle it gently to avoid collision, and do not drag the pipeline on the ground, so as not to damage the probe or scratch the lens.
2. industrial endoscope Before starting the machine, please pay attention to install the battery according to the label. Do not install the battery reversely, so as to avoid short circuit and burn the internal circuit.
3. industrial endoscope After installing the SD card, you need to format the memory card before you can use it. You can select the format of memory card in the menu setting, or operate it through the electric handle.
4.industrial endoscope Insert the front-end pipeline into the tested equipment, operate the rocker to control the direction of the probe, and observe its interior through the display screen. The brightness of the light source can be adjusted to obtain the appropriate illumination, so that the picture can achieve the clearest effect.
5.industrial endoscope If there is oil, water and a lot of dust in the environment, wipe the surface with a dust-free cloth after use to avoid penetration and damage to the lens.
6. It is not allowed to bend the movable joint of the pipeline directly by hand, so as to avoid the breakage of the steel wire traction rope inside the pipeline.
7. Please try not to exceed 70 ° If it must be used in this environment, the time should not exceed 5 minutes.
8. Long term use of this product will make the objective lens of the lens produce stains, resulting in unclear imaging. Please use a dust-free cloth to wipe the stains on the surface of the objective lens before use.
9. When the front end is in bending state, please do not straighten it by hand. Please return the front end by operating the rocker on the host.
10. Do not let the lens in front of the lens collide with hard objects to prevent the lens of the probe from breaking.
11. Do not bend or knot the pipeline, only bend it properly.
12. When using the instrument for testing, if the front end is clamped, do not pull it out by force.
13. The pipeline shall not be stained with water as far as possible. If it needs to be detected in water, please do not soak for more than 10 minutes. After the detection, the surface of the pipeline is attached with water drops, please wipe it clean in time.
14. Please use the original charger to charge so as not to damage the battery.
15. It is forbidden to work under live condition to avoid electric shock.
16. Confirm that the probe of the insertion tube of the industrial endoscope has been in the unlocking state, let it automatically return to the straight position, and then slowly pull it out.
17. Do not look directly at the strong light source emitted by the LED at the end of the endoscope lens at close range to avoid the influence of strong light on vision.
18. It is forbidden to use or store the instrument in places with explosion, strong electromagnetic field and combustible gas, otherwise it may cause fire or explosion.
19. If the inner part of the component is very rough, such as burr, the pipeline and lens may be worn for a long time.
20. If the braided layer on the surface of the pipeline is caught by the burr, the pipeline shall not be pulled by force, resulting in the condition that it cannot be taken out, the pipeline shall be gently rotated to make it separate, and then it shall be taken out carefully.
If you have any other endoscopic problems, welcome to consult Shenzhen micro optoelectronics. We will serve you wholeheartedly. Thank you for your continuous attention to Shenzhen Yuchen Technology Co., Ltd., a professional endoscopic solution service provider!